Fluke A Multimeter DIGITAL | Business & Industrial, Test, Measurement & Inspection, Test Meters & Detectors | eBay!. The listing you’re looking for has ended. Fluke A Multimeter. Condition: For parts or not working. Ended: Jun 08, , AM PDT. Price: US $ Corporation (Fluke) has reproduced this copy of a manual which is no All shipments of JOHN FLUKE MFG. CO. . A A rechargeable Ni Cad battery pack.
|Published (Last):||19 August 2013|
|PDF File Size:||16.50 Mb|
|ePub File Size:||16.47 Mb|
|Price:||Free* [*Free Regsitration Required]|
Image not available Photos not available for this variation. Loosen the two screws securing the circuit board to the top of the front panel. NOTE The fuse holder mA input terminal and fuse should be connected to the positive plug on a GR type connector so that the fuse will be held in the proper place when negative terminal is plugged into the COMMON input terminal This should be done for all inputs to the A mA terminal It offers 20, count resolution in all five functions which provides measurement resolution to 0.
AC Volts Have one to sell? An rms amplitude is that value of alternating current or voltage that results in the same power dissipation in a given resistance as dc current or voltage of the same numerical value. Use the following procedure for operating the A with the transformer: Vary the probe angle and pressure when measuring solid surface temperatures; the highest stable reading will be the most accurate.
Remove all inputs from the A V-fl input. Allow at least 14 hours of recharging a discharge battery pack. To avoid damaging the MOS devices you should 1 minimize handling, 2 discharge personal static before handling the device, 3 handle the MOS device by the body; avoid touching the pins on the device, 4 do not slide the MOS device over any surface, 5 avoid touching plastic, vinyl or styrofoam when working with MOS devices, and 6 always use a grounded soldering iron when installing or removing MOS devices.
Select the DC mA function. No additional import charges at delivery! This procedure may be used as an instrument operational evaluation when the A is being used in locations away from normal calibration equipment.
John Fluke Model A Digital Multimeter | eBay
A description of the control, indicator, or connector is provided in Table 2 – 1. The amplitude of the signal and the time duration of some segments of the waveform are directly related to the input applied to the A. Select the 2 range. Rechargeable battery is present but almost dead Unit turns on without error as shown in picture, then turns itself off right away. Disconnect the input from the V-H input. The A uses flexible connectors for physical and electrical connection of the individual sections of the A1 pcb.
The inputs to be measured are connected to the Analog circuitry via the V-H or mA terminals.
Learn More – opens in a new window or tab International 840a and import charges paid to Pitney Bowes Inc. In the event the part you order has been replaced by a new or improved part, the replacement will be accompanied by an explanatory note and installation instructions, if necessary. Turn R fully clockwise then adjust it counterclockwise until the A display is between 0.
United States and many other countries See details. The A display will indicate 0. This connects the charger to the battery. Timing and Control The rechargeable battery pack, item 7 in Figureis recommended when the A is to be used extensively as a portable multimeter. The formula followed to make this calculation is: The reference voltage is selected, by the Timing and Control circuit, to be opposite in polarity from the output of the Buffer or RMS Converter.
John Fluke Model 8040A Digital Multimeter
See Table for recommended calibration equipment. Adjust C2 for an A display between Special financing available Select PayPal Credit at checkout to have the option to pay over time. Select the appropriate dc voltage range on the Fluke voltmeter.
Back to home page Return to top. Access Information NOTE If the amplitude of the first ms of either oscilloscope display significantly deviates from waveform A3 or B3, U2 is the probable cause of the malfunction. This fuse is located behind the mA input jack and is removed by pressing in lightly on the jack then turning it counterclockwise! The Model 80F division ratio should be checked every month or every year depending on the accuracy requirement.
Learn More – opens in a new window or tab Any international shipping is paid in part to Pitney Bowes Inc.
Therefore, if the measured surface has a low mass e. Reference Designation or Item Number. Similar sponsored items Feedback on our suggestions – Similar sponsored items. The power source typically provides 14 hours of instrument operation. Start of add to list layer. Email to friends Share on Facebook – opens in a new window or tab Share on Twitter – opens in a new window or tab Share on Pinterest – opens in a new window or tab. Required Method for Voltage Source Connection